Toshiba SSA-580A Maintenance
The machine starts normally, but there is no echo when it enters the ultrasonic working interface.
Ultrasound image echo-free is a typical fault that all major brands of ultrasound machines will encounter. Usually, the reason for the failure is that the front-end plate of the power supply of the machine has no high-voltage output or is responsible for the processing function of ultrasound imaging. It is preliminary judged that there may be faults in the A43 FETX board and the A60 ETXR2 board, among which the A43 FETX board is the transmitting and receiving board and the A60 ETXR2 board is responsible for the tube of the high-voltage board. High voltage output of the probe.
After replacing the A60 ETXR2 board, the machine starts up normally. After entering the ultrasonic working interface, the image echo area is restored and the machine runs normally.
Rongtao Medical's mission is to help people realize the efficient circulation utilization of medical resource!
We equipped with:
1.Professional&Passionate pre-sale service team
2.10 years experience repair technician team
3.Reliable&Responsible quality control team
4.Experienced&Specialize after-sale technical support team
Our vision : TO be the top preferred third-party service company worldwide!
If you have any suggestions or question for us.Please contact us.