SIEMENS X300 Maintenance
The fault phenomenon
The X300 machine has a long start-up time, and it works normally when it enters the ultrasonic working interface.
Arrive at the scene to know the situation of the machine to the staff of the department. It takes more than ten minutes for the staff to describe the machine to start each time. In the past, it only took about three minutes for the machine to start normally. Restart the machine for observation, the machine boot process is normal, but in "Checking system configurations..." The interface residence time is long. It is judged that there are some problems in the machine plate, which leads to the prolongation of self-inspection time. After eliminating other faults, it only takes 3 minutes to restart the test machine, and the restart time of the test machine is stable about 3 minutes. It shows that the long start time of the machine is due to the long response time of the control panel circuit board module, and the replacement of the control panel is due to the need for the control panel circuit board module to restart after the first replacement, so the start time is also the same. It will be a little longer.
Exclusion and summary
Replacement of control panel, after repeated testing, boot time is shortened, stable in about 3 minutes, troubleshooting.
Rongtao Medical's mission is to help people realize the efficient circulation utilization of medical resource!
We equipped with:
1.Professional&Passionate pre-sale service team
2.10 years experience repair technician team
3.Reliable&Responsible quality control team
4.Experienced&Specialize after-sale technical support team
Our vision : TO be the top preferred third-party service company worldwide!
If you have any suggestions or question for us.Please contact us.