Repair case: Philips IU22 ultrasound cannot be turned on
The screen of the Philips IU22 ultrasound device stays on the scroll bar interface when it is turned on. After 30 s, a dialog box error code 518 appears. You can choose to enter the settings or directly shut down the dialog box below, but you cannot enter the system operation interface. Restarting the device cannot be solved.
Fault analysis and Troubleshooting
Enter the setting interface, use U disk to export the system error log, analyze it by manufacturer engineer with Philips special software, and determine that there is a problem with the system file. It is recommended to reinstall the system.
Previously, the hard drives removed from scrapped computers and equipment were used for hard drive backups of Philips IU22 and IE33 and other common devices. The IU22 has 3 hard drives, labeled HDD0, HDD1 and HDD2, where HDD0 is the system drive. HDD1 and HDD2 realize image storage.
Replace the system hard disk first, and report 602 error after booting. It is analyzed that the codes of the software and sub-hard disks do not match. After the backup of HDD1 and HDD2 is replaced, the device boots normally, enters the system operation interface, and tests all functions are normal.
When conditions permit, it is important to do a good job of backing up equipment systems and data. At the same time, the hard disk of the scrapped equipment can be used to back up the system and data of commonly used ultrasound equipment, which can effectively use resources. The hard disk capacity of early ultrasound equipment is mostly 500G, and hard disks above 4T may not be recognized. Backup should be performed with 1T or 500G hard disk.
If you have any suggestions or question for us.Please contact us.