Faulty Phenomenon: The 2D image is normal, however there is obvious interference in color blood Image.
Anaysis： Plug the probe in the different connectors. The interface still exits. Check that the probe is normally used on other machines, so it is not the probe problem. It is judged that the BF192 board is faulty.
Troubleshooting：After repair the BF192 board, the image is back to normal.